OpenICE Community Support ForumWelcome to the OpenICE community support forum. Use this forum for submitting bugs, asking for help, and solving problems you are having with OpenICE software. For ideas, general questions, and conversation please use the discussion forum.
I'm having problems to connect PB 840 to a MacBook. I'm using a serial to USB cable. Despite I installed OpenICE and Java, the Device Adapter does not recognizes it, just shows bluetooth port of the Mac. I installed another app called "Serial" and it reads the serial port but I want realtime, high frecuency data to build waveforms and make lung mechanics analysis. What I'm doing wrong?
I am trying to build the mdpn repo but it looks like https://build.openice.info/artifactory is down.
This are the error i get:
* What went wrong:
A problem occurred evaluating project ':data-types:x73-idl-rti-dds'.
> Could not resolve all dependencies for configuration ':data-types:x73-idl-rti-dds:rtiddsgen'.
> Could not resolve org.anarres:cpp-bin:1.2.8-SNAPSHOT.
> Unable to load Maven meta-data from https://build.openice.info/artifactory/libs-snapshot/org/anarres/cpp-bin/1.2.8-SNAPSHOT/maven-metadata.xml.
> Could not GET 'https://build.openice.info/artifactory/libs-snapshot/org/anarres/cpp-bin/1.2.8-SNAPSHOT/maven-metadata.xml'.
> peer not authenticated
> Unable to load Maven meta-data from https://build.openice.info/artifactory/libs-release/org/anarres/cpp-bin/1.2.8-SNAPSHOT/maven-metadata.xml.
> Could not GET 'https://build.openice.info/artifactory/libs-release/org/anarres/cpp-bin/1.2.8-SNAPSHOT/maven-metadata.xml'.
> peer not authenticated
> Unable to load Maven meta-data from https://build.openice.info/artifactory/remote-repos/org/anarres/cpp-bin/1.2.8-SNAPSHOT/maven-metadata.xml.
> Could not GET 'https://build.openice.info/artifactory/remote-repos/org/anarres/cpp-bin/1.2.8-SNAPSHOT/maven-metadata.xml'.
> peer not authenticated
My questions are more related to how you are interfacing with EMRs, and what format you're pushing that data? From a more technical perspective, are you exposing the data through an API? Or are you publishing that data directly to a system? Would love to get a quick overview, from a quasi-technical perspective.
Thanks for your question! The short answer is that EMR communication/integration is done through HL7 (http://www.hl7.org/). Our preferred HL7 version is the development version of HL7 FHIR (http://hl7-fhir.github.io/). If you're interested in using HL7 FHIR and java (what OpenICE uses), a fantastic learning resource is the HAPI FHIR project (http://jamesagnew.github.io/hapi-fhir/).
We have only begun to experiment with EMR integrations, but our current state is best illustrated by the OpenICE Supervisor app 'HL7 Exporter'. You can use the app by installing OpenICE: http://mdpnp.sourceforge.net/. And you can check out the code here: https://sourceforge.net/p/mdpnp/code/ci/master/tree/interop-lab/demo-apps/src/main/java/org/mdpnp/apps/testapp/hl7/. The app will take data from the active patient and export it via HL7 at a user defined time interval.
Our HIMSS15 demo app illustrated some mock "auto-validation" of medical device data that was then exported using the HL7 export app to a fake EMR. That system is representative of some of our future development ideas that will be on our roadmap eventually. For now, there is still a lot of work to do in this area...
Feel free to continue the conversation. Thanks,
I tried build your mdpnp project(https://github.com/mdpnp/mdpnp), but it occurs error like below.
It looks your server has some problem(502 error occurs), can you check out this?
I restarted the server, please try again.
I would like to use Android App,but I can not build it.
NOTE: To build with OpenSplice compatibility please set the OSPL_HOME and SPLICE_TARGET environment variables.
How could I build it?
The "android-apps" project was a simple demonstration of connecting directly to the Nonin Bluetooth Pulse Oximeter from an Android device. The project is no longer part of the build since 19-June-2013. It remains in the repo for reference but it has been deprecated since that time.
I has try install OpenICE and successful installed.
I has create simulated device and link to patient (list in patient ID), it look working fine.
I has run HL7 export and start it, but HL7 message has been capture did not show active patient data , why this thing happened?
hope you can help me. thank you
C [libnddscore.so+0x72d48] call_gmon_start+0x7
I hope you could give me some advice.
I've been attempting to connect to our IntelliVue MX800 over LAN with the OpenICE demo apps.
I have established a network connection to the monitor from my laptop successfully, and can ping the device. However, I'm unable to get the device adapter to connect to the monitor over LAN.
I inspected network traffic with Wireshark, and saw no communication at all addressed to the monitor IP that I entered when starting the device adapter.
Customer support service by UserEcho